# What is the limitations of Kelvin Double Bridge?

## What is the limitations of Kelvin Double Bridge?

Limitations of Kelvins Bridge The sensitive galvanometer is used for detecting the balance condition. The high measurement current is required for obtaining the good sensitivity.

## What is purpose of reverse switch in Kelvin Double Bridge?

Current Reversing Switch : A battery reversing switch is provided to reverse the current in the bridge to eliminate thermal effects.

**Why Kelvin Bridge is used for measurement of low resistance Mcq?**

7. Why Kelvin bridge is used for measurement of low resistance? Explanation: While measuring very low resistances the contact and lead resistances cause significant errors in the value of the measured resistance. As a result Kelvin bridge is used for measurement of low resistances.

**How many four terminal resistance are there in a Kelvin double bridge?**

Any contact resistance at the current lead-in terminals is also reduced by the new points. Additional information: Circuit Diagram: The kelvin double bridge has four terminal resistors for the low resistance arms and the second set of ratio arms (thus the name “double bridge”).

### What are the application of Kelvin Bridge?

A Kelvin bridge, also called a Kelvin double bridge and in some countries a Thomson bridge, is a measuring instrument used to measure unknown electrical resistors below 1 ohm. It is specifically designed to measure resistors that are constructed as four terminal resistors.

### Why can’t a Kelvin bridge be used for the?

Explanation: A Kelvin bridge can be used for the measurement of high Quality factor values. Due to bridge imbalance problem, Kelvin bridge can’t be used for measuring low Quality factor values.

**What is Kelvin method?**

Hint:Kelvin’s method determines the resistance of the galvanometer using the principle of Wheatstone bridge, which makes it easier to find the resistance since you only need to know the ratio of the lengths and not the specific value of resistance of wires.

**How is bridge balanced Mcq?**

Explanation: The bridge circuit is said to be balanced when no current flows through the null indicator which is generally a galvanometer. Explanation: During balance condition or equilibrium there exists a certain relationship between the component values of the four arms of the bridge.

#### What is the balance equation for Kelvin Bridge?

The ratio p/q is made equal to P/Q. Under balance conditions there is no current through the galvanometer which means that the voltage drop between a and b, Eab is equal to voltage drops Eamd between a and c. Eq (2) is the usual working equation for the kelvin bridge.

#### What are the advantages of Kelvin double bridge over potentiometer?

Because by using Kelvin’s Double Bridge, your end result i.e. the value of unknown resistance will not be influenced by resistances of connecting lead.

**What is Kelvin contact?**

A pair of contacts that is designed to connect a force-and-sense pair to a single terminal or lead simultaneously is called a Kelvin contact. A clip, often a crocodile clip, that connects a force-and-sense pair (typically one to each jaw) is called a Kelvin clip.

**What is balance point in Kelvin method?**

Then the jockey slides over the meter bridge points at the two extreme edges, and checks if the galvanometer shows deflection on both sides. Now, the jockey is slid to such a point, where the galvanometer shows no deflection. This point is called the balance point.

## What is the accuracy of Kelvin double bridge?

What is the accuracy of Kelvin Double Bridge? The unknown resistance value can be obtained from the two arms of Kelvin double bridge, typically, it measures 1- 0.00001 ohm with an accuracy ± 0.05% to ±0.2%. A bridge is an electrical circuit, which is used in Labourites for measuring various parameters.

## What is the purpose of the Kelvin bridge?

Kelvin Bridge. Definition: The Kelvin bridge or Thompson bridge is used for measuring the unknown resistances having a value less than 1Ω. It is the modified form of the Wheatstone Bridge. Wheatstone bridge use for measuring the resistance from a few ohms to several kilo-ohms.

**How are ratio arms used in Kelvin bridge circuit?**

In this the ratio arms p and q are used to connect the galvanometer at the correct point between j and k to remove the effect of connecting lead of electrical resistance t. Under balance condition voltage drop between a and b (i.e. E) is equal to F (voltage drop between a and c)

**Is the Kelvin bridge independent of load and contact resistance?**

There is no effect of load and contact resistance in the Kelvin bridge since the bridge is independent of load and contact resistance. 7). What is the accuracy of Kelvin Double Bridge?